C16/D16 Calibration "T"
Calibration bypass T for the C16/D16. (00-1088)
Bypass calibration T for the C16/D16 PortaSense detector.
When Supplying gas to the device from a pressurized source requires this bypass-T to be inserted between the source and the manifold (Figure below). This is required to limit gas pressure on the intake side of the sample pump when the source is open, and prevent the inlet from being blocked when the source is closed. The flow rate into the T must exceed the pump flow rate of 400cc/minute, or the calibration gas may become diluted. A flow rate of 500 cc/min into the T is recommended.
|Call For Price||No|
|Monitor Type||C16, D16|
Write Your Own Review